Definition:Jacketed Glass Reactor
is designed double-level glass. The reaction material is put into the inner, which can be stirred. In the interbed with different cold and hot medium (refrigerating fluid, hot water and hot oil), do some heat circulations and cooling reactions. Under the constant temperature and ordinary pressure, as well as negative pressure, stirring reaction can be done in seal Glass Reactor
, also back-flow, distillation etc.
* lab glass reactor Function:
1. Vacuum stirring reaction
2. High and low-temperature reaction
3. Constant speed running mixing reaction
4. Distillation, refluence and condensation
5. Negative pressure operation and separation
6. Rectify through assembling the rectifying pillar
7. Assembly of the production line in accordance of specific demands.
* Chemical double glass reactor Features:
1. GG17 high-borosilicate glass with nice physical-chemical properties; experiment temperature -100 to 400 degree C, Resistant to strong acidic properties of materials; only touching with glass parts & PTFE, no cross-contamination.
2. Double cylinder; kettle cover is separated from kettle body. It is convenient to disassemble, clean and install.
3. Stirring adopts to mechanical sealing, it keeps higher vacuum degree among the same products in the market. It also keeps high vacuum degree when stirring.
4. Glass(PTFE)discharge valve, non-dead angle mixing, fast emptying, no residue.
5. Frequency control speed gear motor, smooth running, high torque, automatic booster, no spark, noise free, long life.
6. SUS frame and cast aluminum material kettle body support, elegant appearance, stable structure.
7. Circulating water/oil bath, high and low temperature machine can supply heating source; Cooling machine: Cooling Chiller
, high and low temperature machine, convenient and practical.
8. Scientific & innovative structure, practical, elegant appearance, domestic industry-leading.
9. Explosion proof is optional.
1. Keep every glass part clean before installation.
2. Each standard port, milling mouth and sealing surfaces should be coated with a small amount of vacuum grease, to keep good sealing.
3. Please read the manual carefully before installation, use, maintenance.
4. Your voltage should be in consistent with the provisions of this instrument.
5. No load is not suitable for high speeds.
6. The ambient temperature and humidity have big effects on the the electrical box life. Please keep the governor and motor dry.
7. The installation and use should be operated by professional operators.
• Chemical reactions, crystallizations, work-ups, distillations, bioreactions etc.
• Process optimization, process development or production
• Pharmaceutical, petrochemistry, food, flavour and fragrance,fine chemical synthesis, agrochemistry, paints and dyes etc.